Remove and retain the two screws that hold the sensors board in place. When triggered by a sensor enable pulse from the main processor, Q10 turns on so that DCSW, limited by R28 and R29, is applied across the sensor emitters. Both new and second hand devices will be sold with a warranty, the length of which will be determined by the equipment type and agreed at the point of sale. The pump provides a purge facility to assist with this process. May be opened and closed without it. We hope that you will be very happy with your purchase however if you are not, please contact us as soon as you recognise the problem. Mobile Data Power Model:
|Date Added:||2 December 2011|
|File Size:||39.21 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Smiths Medical recommend that the batteries are checked at least annually see page Issue 7 August Service Manual 2 7. Syringe stiction Stiction for a syringe hraseby from brand to brand as well as from batch to batch.
Manuel d installation Installation manual.
” + heading + “
Wheel speed sensors in motor vehicles. Remove the Status 3030 board see page Add to Watch list Watching. To renew a blown fuse, the pump casing must be disassembled see page to access all three fuses.
Pins 2 and 3 on the main processor control its mode of operation. If the item is subject to Health Canada or FDA regulation and it is to be used for regulated medical services in the Canada or the USA, your payment for it indicates you are an authorized purchaser and will comply with these regulations.
Cover not closed A flag is fixed to the inside of the left hinge of the syringe cover.
Our Ideas, Graseeby Success. Trademark Information is a registered trademark of. Remove the faulty clutch and disc assembly. This item is sold as fully functional.
Taking care not to trap any leads, reassemble the casing 4. Operating Instructions Ultrasound Technologies PD1v vascular pocket doppler Operating Instructions Before using your Pocket Doppler for the first time, please read these operating instructions carefully.
Sign In or Register. The pump can also display the total mass and volume of drug infused since the totalisers were last reset; the number of PCA demands; and the number of good successful PCA demands. Working from the top downwards and using light pressure lay the panel into the case recess.
Simply complete the form below and click the ‘Send It’ button. It is essential that clinical staff remain within visual and audible range of the pump so that critical alarms can be seen or heard and responded to. A large text vacuum fluorescent display was added, and the increased syringe size range of the remained. More than 8 hours of normal pump operation when the batteries are fully charged.
Access to added-value services including medical equipment Test and Calibration, CQC advice and health and safety audits. Remove and retain the Spirol pin which will now be accessible. Buy it now – Add to Watch list Added to your Watch list. Seller assumes all responsibility for this listing. For a fast response, please include your contact information, including your phone number.
Modify the syringe by sawing off the pointed end.
Locate the lock housing in place and tap home the Spirol fixing pin. Disconnect the two bottom left hand connectors PL11 and PL12 from the board by prising the side retaining clip from each connector and then pulling the connector out.
The left hand screw is 25 mm long and the right hand screw is 8 mm long.
If necessary, stick a new adhesive voltage rating label over the existing label on the underside of the pump. To avoid over- or under- infusion, always verify that the brand and size of the loaded syringe are the 33300 as the brand and size displayed on the screen before starting an infusion.
STAT or 1 to 15 minutes.